Newest Original TP Touch Screen for AUTEL MaxiSYS MS906 Auto Diagnostic Scanner Free Shipping
Original TP Touch Screen for AUTEL MaxiSYS MS906 Auto Diagnostic Scanner
This is an original touch screen used for AUTEL MaxiSYS MS906 system.
1.Please confirm your MS906 touch screen cannot work any more before your purchase. Better check it with our technicians.
2.You should be able to install the screen yourself, as we cannot offer this tech support for you.
3.No one but yourself should take responsibility if you break something.
Touch screen exchange tips
Tip: if your touch screen is the same version as this one, please kindly note this step:
when you exchange the touch screen, at the same time, you need to dismantle R375 resistance on the PCB.
1pc x Original TP Touch Screen for AUTEL MaxiSYS MS906 Auto Diagnostic Scanner
If you have any problem, please do not hesitate to contact us. We respond to our email 7/24.
Important Notes on Replacement of New Supplier’s TP Screen on MaxiSys Tablets
Important Notes on Replacement of New Supplier's TP Screen on MaxiSys Tablets
A. Part Number of new TP screen Involved
2. Remove J11, J12 buttons (as showed below) for MY906
d. Pasting way of TP for MY906,MY906BT,MY906TS
TP plate should be pasted parallelly with the magnesium alloy stent.
The above new TP screen is provided by a new supplier, which will be used for TP replacement of the above models.
a. Update operating system to the latest version first
E.g. Maxisys 906 V03.46.00, Maxisys906BT V03.31.00 , MY908 V03.41.00
If the MaxiSys was out of update period, please use reflashing program(sent via Wetransfer) to update it to the free version.
b. Remove Resistance
1. Remove R289 Resistance for MY908
2. Remove R375 Resistance for MY906/MY906BT
c. Remove specific buttons for MY906,MY906BT,MY906TS
1. Remove J11, J12 buttons(as showed below) for MY06BT, MY906TS